**SMD-5TR** is our **Trinocular Zoom 7X – 45X Stereo Microscope package with Incident Ring Light LED on a Boom Stand for Surface Mount Device Inspection. **The Trinocular tube gives the microscope the ability to accept SLR / HDMI /USB 2.0 / USB 3.0/ Analog C” mountable camera’s with the proper adapter. This will allow the engineer to incorporate observation, image capture of still and live video and measurement capabilities while providing an on-screen interface for viewing real time live images. This is Meiji Techno America’s complete turnkey microscope package, based on our EMZ Series microscopes. Those who must work to the most demanding standards of the SMD industries require a microscope system for assembly, repair, inspection and Quality Control. The SMD Inspection Series is equipped with Meiji Techno’s most versatile Boom Stand which can be adjusted vertically or horizontally and side to side to address the ergonomic needs of every production or test engineer. Equipped with an auxiliary lens which gives the user over 6 inches of working distance. This allows ample working room to allow the engineers to solder and work on PCB boards being inspected without compromising Ergonomics.

**Included:**

- •EMZ-5TR *Trinocular Zoom Stereo microscope body with zoom range, W.D. 93mm magnification from 7X to 90X (Using SWF10X and SWF20X eyepieces)

- •MA502 *SWF10X super widefield eyepiece (Paired), F.N. 23

- •MA504 *Super Widefield Eyepiece 20X (paired) (With 25mm reticle mount)
- •S-4100 *Universal boom stand with 20 mm drop down post for F or FX block 400mm vertical post

- •F *Focus block with Coarse Focus fits 20mm pole stand, 84.5mm Inner Diameter

- •MA517 *Supplementary lens 0.5X, working distance 150mm (5.9”)

- •MA964 *LED Ring illuminator with automatic voltage sensing power supply, 115V or 240V 60 LED Array, 10,000 hour rated lifetime, CE approved, fits 53mm-60mm nosepiece Wattage: 10W. Transformer case: Length 150mm, width 60mm, height 83mm

**EMZ-5TR SPECIFICATIONS**

**Body:** Trinocular, Rotatable 360°, Cast Aluminum Alloy Construction

**Zoom Ratio:** 6.5:1

**Zoom Range:** 0.7x - 4.5x

**Zoom Knobs Markings:** 0.7, 1, 1.5, 2, 2.5, 3, 3.5, 4, 4.5

**Zoom Control:** Dual, graduated, bilateral

**Magnification:** 7x - 45x (10x eyepieces)

**Objective Lens:** Brass Construction

**Field of View:** 32mm - 5.1mm

**Working distance:** 93mm

**Eyetube Inclination:** 45°

**Eyepieces: **5x, 7.5x, 10x, 12.5x, 15x, 20x, and 30x (optional)

**Interpupillary Distance:** 54mm - 75mm

**Dioptric Adjustment:** Dual (+/- 5 diopters)

**Extended Mag Range:** 2.1x - 270x (With auxiliary lenses and eyepieces)

**Extended Working Distances:** 34mm - 251mm (With auxiliary lenses and eyepieces)

**Auxiliary Lenses: **0.3x (252mm), 0.44x (181mm), 0.5x (150mm), 0.75x (97mm), 1.5x (49mm), and 2x (34mm) (optional)

**Mounting:** Fits 84.5mm Diameter Focus Holders

**Manufacturer:** MADE IN JAPAN

**Warranty:** LIMITED LIFETIME WARRANTY

**MA502***SWF10X super widefield eyepiece (Paired), F.N. 23**MA504***Super Widefield Eyepiece 20X (paired) (With 25mm reticle mount)**S-4100***Universal boom stand with 20 mm drop down post for F or FX block 400mm vertical post**MA517***Supplementary lens 0.5X, working distance 150mm (5.9”)**MA964***LED Ring illuminator with automatic voltage sensing power supply, 115V or 240V 60 LED Array, 10,000 hour rated lifetime, CE approved, fits 53mm-60mm

**Includes:**

•**EMZ-5TR** *Trinocular Zoom Stereo microscope body with zoom range, W.D. 93mm magnification from 7X to 90X (Using SWF10X and SWF20X eyepieces)

•**MA502** *SWF10X super widefield eyepiece (Paired), F.N. 23

•**MA504** *Super Widefield Eyepiece 20X (paired) (With 25mm reticle mount)

•**S-4100** *Universal boom stand with 20 mm drop down post for F or FX block 400mm vertical post

•**MA517** *Supplementary lens 0.5X, working distance 150mm (5.9”)

•**MA964** *LED Ring illuminator with automatic voltage sensing power supply, 115V or 240V 60 LED Array, 10,000 hour rated lifetime, CE approved, fits 53mm-60mm nosepiece Wattage: 10W. Transformer case: Length 150mm, width 60mm, height 83mm

•**Meiji Techno Dust Cover (MA703)**

•**Rubber Eyeguards/ Eyeshields (Paired) (MA600)**

•**EMZ Meiji Techno Instruction Manual**

•**Limited Lifetime Warranty Card**

**Product Information:****Manufactured :** 100% MADE IN JAPAN**Warranty: ** LIMITED LIFETIME WARRANTY**Electrical Warranty:** 1 YEAR**Customer Support:** 1-(800)-832-0060 ( U.S.A. Based )

**Package Information:****Dimensions:** Length: 650mm x Width 265mm x Height 400mm (Footprint)**BOX 1** (S-4100)**Weight:** 45 Lbs.**Box Size:** 25” x 14” x 9”**BOX 2** (Head , Eyepieces , Auxilliary Lens, Illumination)**Weight: ** 12 Lbs.**Box Size:** 15” x 13” x 10”

**MDM (MEDICAL DEVICE MANUFACTURING) SERIES ****BOOM STAND INSPECTION SYSTEM ZOOM STEREO MICROSCOPES**

**NOTE: Surface-Mount Device (SMD) and Surface-Mount Technology (SMT) and Medical Device Manufacturing Series Boom Stand Zoom Stereo Microscope Inspection System is a complete microscope system comprised of: **

• Stereo Zoom Microscope Body (Binocular or Trinocular Zoom Stereo Microscope)

• Eyepieces (Paired Super Wide Field 10X eyepiece – accepts a 25mm reticle)

• Boom Stand (Provides complete flexibility for horizontal and vertical adjustment, sturdy metal 40 lbs base for excellent stability with 20mm drop down post

•F focusing Holder (Smooth precise focusing over 55mm of travel, with slip clutch and tension adjustment. Coarse focus focusing block with 84.2mm diameter opening accepts EMZ series)

•Auxilliary Lens (MA517 –W.D. 150mm or MA794- W.D. 174mm)

•MA964 or FT192 or FT19120

SMT & SMD and Medical Device Manufacturing Boom Stand Zoom Stereo Microscope Inspection system is Meiji Techno America’s complete turnkey microscope package, based on our EMZ Series microscopes. Those who must work to the most demanding standards of the SMT and SMD industries and Medical Device Manufacturing (MDM) facilities require a microscope system for assembly, repair, inspection and Quality Control.

The SMT & SMD Inspection Series and Medical Device Manufacturing Series is equipped with Meiji Techno’s most versatile Boom Stand which can be adjusted vertically or horizontally and side to side to address the ergonomic needs of every production or test engineer. Equipped with an auxiliary lens which gives the user over 6 inches of working distance. This allows ample working room to allow the engineers to solder and work on PCB boards and inspect Medical Stents or any Medical Device being inspected without compromising Ergonomics.

**Surface-mount Technology (SMT)** is a method for producing electronic circuits in which the components are mounted or placed directly on to the surface of printed circuit boards

(PCB’s). An electronic device so made is called **Surface-mount Device (SMD)**. In the industry it has largely replaced the through-hole technology construction method of fitting components with wire leads into holes in the circuit board. Since both technologies can be used on the same board, with the through-hole technology used for components not suitable for surface mounting such as large transformers and heat-sinked power semiconductors. SMT component is usually smaller than its through hole counterpart because it has either smaller leads or no leads at all. It may have short pins or leads of various styles, flat contacts, or a matrix of solder balls (BGA’s) or termination on the body of the component.

Meiji Techno America offers a variety of illuminations as an option because where components are to be placed, the printed circuit board normally has flat, usually tin-lead silver or gold plated copper pads without holes, called solder pads. Our Inspection Series allows the boards to be visually inspected for missing or misaligned components and solder bridging, if needed, they are sent to a rework station where a Meiji Techno Microscope system is used to inspect errors and repaired by a human operator . They are sent to the testing stations (in-circuit testing and/or functional testing) to verify that they operate correctly. Defective surface-mount components can be repaired by using our stereo zoom systems by using soldering irons (for some connections) , or using a non contact rework system. In most cases a rework system is the better choice because SMD work with a soldering iron requires and excellent Optical Microscope quality and considerable skill.

**Medical Device Manufacturing (MDM**) requires accurately documenting, inspecting and analyzing information accurately and efficiently in your R&D division. Meiji Techno America’s medical Device systems allow you to have a Japanese quality Optical system to help in aid in your next life saving technology.

Since the Medical Device Industry is a highly regulated and stringent in quality control and assurance and failure analysis it is critical for a test engineer to be able to see and validate the product without fail. Medicated Stents, catheters, guidewires, Angio balloons , neurological stents requires efficient assembly, inspection and documentation with the highest level of accuracy. Meiji Techno’s Boom Inspection system improves production throughput especially with repetitive tasks by reducing the technician’s eye strain with ergonomics in mind to aid in high precision inspection. Doctors are always trying to come up with new ways to prevent arteries from re-closing after an angioplasty or stent procedure. In recent years, doctors have used new types of stents. Some of these are coated with medicines that help keep the vessel from re-closing. These coated stents slowly release their medicine into the tissue around the stent, slowing down or stopping restenosis. Coated stents have shown promise for improving the long-term success of this procedure.

Researchers are also designing smaller stents and custom-designed stents, which will fit into smaller vessels. They are also designing stents for vessels that have many blockages, including vessels with side branches. Other stents are being coated with blood-thinning medicines to prevent blood clots from forming within the stent. Most recently, researchers have begun developing non-metal stents that will dissolve into the artery over time. These stents open the artery, giving the same results as metal stents, but reduce the risk of restenosis because they do not stay in the artery.

Meiji Techno America offers a large variety of configured and customizable solutions to meet your imaging needs, tailored solutions for your budget and application. Enhanced ergonomics and worker comfort for Medical Device Inspection and re-work achieves a higher throughput and better workflow with comfort and accuracy in mind for higher efficiency.

Model |
Zoom Range |
Auxiliary Lens |
Working Distance (mm) |
SWF10X Eyepieces |
SWF12.5X Eyepieces |
SWF15X Eyepieces |
SWF20X Eyepieces |
SWF30X Eyepieces |
|||||

Total Mag. |
Field (mm) |
Total Mag. |
Field Diameter (mm) |
Total Mag. |
Field Diameter (mm) |
Total Mag. |
Field Diameter (mm) |
Total Mag. |
Field Diameter (mm) |
||||

EMZ-1 |
1X -3X | none | 93 | 10X ~ 30X | 23 ~ 7.7 | 12.5X ~ 37.5x | 20 ~ 6.6 | 15X ~ 45X | 15.4 ~ 5.1 | 20X ~ 60X | 11.5 ~ 3.8 | 30X ~ 90X | 7.7 ~ 2.5 |

0.3X | 251 | 3X ~ 9X | 76 ~ 25.6 | 3.75X ~ 11.25X | 66 ~ 22 | 4.5X ~ 13.5X | 51.3 ~ 17 | 6X ~ 18X | 38 ~ 12.6 | 9X ~ 27X | 25.5 ~ 8.3 | ||

0.44X | 181 | 4.4X ~ 13.2X | 52 ~ 17.5 | 5.5X ~ 16.5X | 45 ~ 15 | 6.6X ~ 19.8X | 35 ~ 11.5 | 8.8X ~ 26.4X | 26 ~ 8.6 | 13.2X ~ 39.6X | 17.5 ~ 5.6 | ||

0.5X | 150 | 5X ~ 15X | 46 ~ 15.3 | 6.25X ~ 18.75X | 40 ~ 13.2 | 7.5X ~ 22.5X | 30.6 ~ 10.2 | 10X ~ 30X | 23 ~ 7.6 | 15X ~ 45X | 15.4 ~ 5 | ||

0.75X | 97 | 7.5X ~ 22.5X | 30.6 ~ 10.2 | 9.37X ~ 28.1X | 26.6 ~ 8.8 | 11.25X ~ 33.75X | 20.5 ~ 6.8 | 15 ~ 45X | 15 ~ 5.0 | 22.5X ~ 67.5X | 10.2 ~ 3.3 | ||

1.5X | 49 | 15X ~ 45X | 15.3 ~ 5.1 | 18.75X ~ 56.25X | 13.3 ~ 4.4 | 22.5X ~ 67.5X | 10.2 ~ 3.4 | 30X ~ 90X | 7.7 ~ 2.5 | 45X ~ 135X | 5.1 ~ 1.6 | ||

2.0X | 34 | 20X ~ 60X | 11.5 ~ 3.8 | 25X ~ 75X | 10 ~ 3.3 | 30X ~ 90X | 7.7 ~ 2.6 | 40X ~ 120X | 5.7 ~ 1.9 | 60X ~ 180X | 3.8 ~ 1.2 | ||

EMZ-2 EMZ-2TR EMZ-5 EMZ-5TR EMZ-9 |
0.7X~ 4.5X |
none | 93 | 7X ~ 45X | 32.8 ~ 5.1 | 8.75X ~ 56.25X | 28.5 ~ 4.4 | 10.5X ~ 67.5X | 22 ~ 3.4 | 14X ~ 90X | 16.4 ~ 2.5 | 21X ~ 135X | 11 ~ 1.7 |

0.3X | 251 | 2.1X ~ 13.5X | 109 ~ 17 | 2.62X ~ 16.87X | 95 ~ 14.6 | 3.15X ~ 20.25X | 73 ~ 11.4 | 4.2X ~ 27X | 54.7 ~ 8.5 | 6.3X ~ 40.5X | 36.6 ~ 5.7 | ||

0.44X | 171 | 3.08X ~ 19.8X | 74.5 ~ 11.5 | 3.85X ~ 24.75X | 64.7 ~ 10 | 4.62X ~ 29.7X | 50 ~ 7.7 | 6.16X ~ 39.6X | 37.2 ~ 5.9 | 9.24X ~ 59.4X | 25 ~ 3.8 | ||

0.5X | 148 | 3.5X ~ 22.5X | 65.7 ~ 10.2 | 4.37X ~ 28.12X | 57 ~ 8.8 | 5.25X ~ 33.75X | 44 ~ 6.8 | 7X ~ 45X | 32.8 ~ 5.1 | 10.5X ~ 67.5X | 22 ~ 3.4 | ||

0.75X | 97 | 5.25X ~ 33.75X | 43.8 ~ 6.8 | 6.56X ~ 42.18X | 38 ~ 5.8 | 7.87X ~ 50.62X | 29.3 ~ 4.5 | 10.5X ~ 67.5X | 21.9 ~ 3.4 | 15.75X ~ 101.25X | 14.6 ~ 2.2 | ||

1.5X | 49 | 10.5X ~ 67.5X | 21.9 ~ 3.4 | 13.12X ~ 84.37X | 19 ~ 2.9 | 15.75X ~ 101.25X | 14.6 ~ 2.2 | 21X ~ 135X | 10.9 ~ 1.7 | 31.5X ~ 202.5X | 7.3 ~ 1.1 | ||

2.0X | 33 | 14X ~ 90X | 16.4 ~ 2.5 | 17.5x ~ 112.5X | 14.2 ~ 2.2 | 21X ~ 135X | 11 ~ 1.7 | 28X ~ 180X | 8.2 ~ 1.2 | 42X ~ 270X | 5.5 ~ 0.8 | ||

EMZ-6 |
0.62X~ 4.2X | none | 108 | 6.5X ~ 42X | 35 ~ 5.4 | 8.12X ~ 52.5X | 30.7 ~ 4.7 | 9.75X ~ 63X | 23.7 ~ 3.6 | 13x ~ 84X | 17.7 ~ 2.7 | 19.5X ~ 126X | 11.8 ~ 1.8 |

0.5X | 180 | 3.25X ~ 21X | 70 ~ 10.8 | 4.0X ~ 26.2X | 61.4 ~ 9.4 | 4.87X ~ 31.5X | 47.4 ~ 7.2 | 6.5X ~ 42X | 35.4 ~ 5.4 | 9.75X ~ 63X | 23.6 ~ 3.6 | ||

Model |
Zoom Range |
Auxiliary Lens |
Working Distance (mm) |
SWF10X Eyepieces |
SWF12.5X Eyepieces |
SWF15X Eyepieces |
SWF20X Eyepieces |
SWF30X Eyepieces |
|||||

Total Mag. |
Field Diameter (mm) |
Total Mag. |
Field Diameter (mm) |
Total Mag. |
Field Diameter (mm) |
Total Mag. |
Field Diameter (mm) |
Total Mag. |
Field Diameter (mm) |
||||

EMZ-10 Z-7100 |
0.7X~ 4.5X | none | 110 | 7X ~ 45X | 32.8 ~ 5.1 | 8.75X ~ 56.25X | 28.5 ~ 4.4 | 10.5X ~ 67.5X | 22 ~ 3.4 | 14X ~ 90X | 16.4 ~ 2.6 | 21X ~ 135X | 11 ~ 1.7 |

0.3X | 326 | 2.1X ~ 13.5X | 109 ~ 17 | 2.62X ~ 16.87X | 95 ~ 14.6 | 3.15X ~ 20.25X | 73 ~ 11.3 | 4.2X ~ 27X | 54.6 ~ 8.6 | 6.3X ~ 40.5X | 36.6 ~ 5.7 | ||

0.44X | 247 | 3.08X ~ 19.8X | 74.5 ~ 11.5 | 3.85X ~ 24.75X | 64.7 ~ 10 | 4.62X ~ 29.7X | 50 ~ 7.7 | 6.16X ~ 39.6X | 37.2 ~ 5.9 | 9.24X ~ 59.4X | 25 ~ 3.8 | ||

0.5X | 194 | 3.5X ~ 22.5X | 65.6 ~ 10.2 | 4.37X ~ 28.12X | 57 ~ 8.8 | 5.25X ~ 33.75X | 44 ~ 6.8 | 7X ~ 45X | 32.8 ~ 5.1 | 10.5X ~ 67.5X | 22 ~ 3.4 | ||

0.75X | 127 | 5.25X ~ 33.75X | 43.8 ~ 6.8 | 6.56X ~ 42.18X | 38 ~ 5.8 | 7.87X ~ 50.62X | 29.3 ~ 4.5 | 10.5X ~ 67.5X | 21.8 ~ 3.4 | 15.75X ~ 101.25X | 14.6 ~ 2.2 | ||

1.5X | 64 | 10.5X ~ 67.5X | 21.8 ~ 3.4 | 13.12X ~ 84.37X | 19 ~ 2.9 | 15.75X ~ 101.25X | 14.6 ~ 2.2 | 21X ~ 135X | 10.9 ~ 1.7 | 31.5X ~ 202.5X | 7.3 ~ 1.1 | ||

2.0X | 44 | 14X ~ 90X | 16.4 ~ 2.5 | 17.5X ~ 112.5X | 14.2 ~ 2.2 | 21X ~ 135X | 11 ~ 1.7 | 28X ~ 180X | 8.2 ~ 1.2 | 42X ~ 270X | 5.5 ~ 0.8 | ||

EMZ-12 EMZ-12TR |
0.4X~ 2.5X | none | 185 | 4X ~ 25X | 57 ~ 9.2 | 5X ~ 31.2X | 50 ~ 8 | 6X ~ 37.5X | 38.5 ~ 6.1 | 8X ~ 50X | 28.7 ~ 4.6 | 12X ~ 75X | 19.2 ~ 3 |

0.7X | 233 | 2.8X ~ 17.5X | 82.1 ~ 13.1 | 3.5X ~ 21.8X | 71.4 ~ 11.4 | 4.2X ~ 26.2X | 55 ~ 8.8 | 5.6X ~ 35X | 41 ~ 6.5 | 8.4X ~ 46.2X | 27.5 ~ 4.4 | ||

EMZ-13 EMZ-13TR EMZ-13VX |
1X~ 7.0X | none | 90 | 10X ~ 70X | 22.0 ~ 3.14 | 12.5X ~ 87.5X | 17.6 ~ 2.51 | 15X ~ 105X | 14.66 ~ 2.09 | 20X ~ 140X | 11.0 ~ 1.57 | 30X ~ 210X | 7.33 ~ 1.04 |

0.3X | 230 | 3X ~ 21X | 73.33 ~ 10.47 | 3.75X ~ 26.25X | 58.6 ~ 8.38 | 4.5X ~ 31.5X | 48.88 ~ 6.98 | 6.0X ~ 42.0X | 36.6 ~ 5.23 | 9.0X ~ 63.0X | 24.4 ~ 3.49 | ||

0.44X | 167 | 4.4X ~ 30.8X | 50.0 ~ 7.13 | 5.5X ~ 38.5X | 40 ~ 4.5 | 6.6X ~ 46.2X | 33.1 ~ 4.5 | 8.8X ~ 61.6 | 25.0 ~ 3.5 | 13.2X ~ 92.4X | 16.5 ~ 2.3 | ||

0.5X | 139 | 5X ~ 35X | 44.0 ~ 6.28 | 6.25X ~ 43.75X | 35.2 ~ 5.02 | 7.5X ~ 52.5X | 29.3 ~ 4.19 | 10X ~ 70X | 22.0 ~ 3.14 | 15X ~ 105X | 14.6 ~ 2.09 | ||

0.75X | 92 | 7.5X ~ 52.5X | 29.3 ~ 4.19 | 9.37X ~ 65.62X | 23.47 ~ 3.35 | 11.25X ~ 78.75X | 19.55 ~ 2.79 | 15X ~ 105X | 14.66 ~ 2.09 | 22.5X ~ 157.5X | 9.77 ~ 1.39 | ||

1.5X | 48 | 15X ~ 105X | 14.6 ~ 2.09 | 18.75X ~ 131.25X | 11.73 ~ 1.67 | 22.5X ~ 157.5X | 9.77 ~ 1.39 | 30X ~ 210X | 7.33 ~ 1.04 | 45X ~ 315X | 4.88 ~ 0.69 | ||

2.0X | 32 | 20X ~ 140X | 11.0 ~ 1.57 | 25X ~ 175X | 8.8 ~ 1.25 | 30X ~ 210X | 7.33 ~ 1.04 | 40x ~ 280x | 5.5 ~ 0.78 | 60X ~ 420X | 3.66 ~ 0.52 | ||

Model |
Zoom Range |
Auxiliary Lens |
Working Distance (mm) |
SWF10X Eyepieces |
SWF12.5X Eyepieces |
SWF15X Eyepieces |
SWF20X Eyepieces |
SWF30X Eyepieces |
|||||

Total Mag. |
Field Diameter (mm) |
Total Mag. |
Field Diameter (mm) |
Total Mag. |
Field Diameter (mm) |
Total Mag. |
Field Diameter (mm) |
Total Mag. |
Field Diameter (mm) |
||||

EMZ-8TR EMZ-8TRD EMZ-8U EMZ-8UD EMZ-8 |
0.7X~ 4.5X | none | 104 | 7X ~ 45X | 32.8 ~ 5.1 | 8.75X ~ 56.25X | 28.5 ~ 4.4 | 10.5X ~ 67.5X | 22 ~ 3.4 | 14X ~ 90X | 16.4 ~ 2.6 | 21X ~ 135X | 11 ~ 1.7 |

0.28X | 300 |
1.96X ~ 12.6X |
112.3 ~ 17.4 | 2.45X ~ 15.75X | 97.5 ~ 15.3 | 2.94X ~ 18.9X | 78.1 ~ 12.2 | 3.92X ~ 25.2X | 55.6 ~ 8.7 | 5.88X ~ 37.8X | 34 ~ 5.5 | ||

0.35X | 250 | 2.45X ~ 15.75X | 93.7 ~ 14.5 | 3.06X ~ 19.68X | 81.4 ~ 12.5 | 3.67X ~ 23.62X | 62.8 ~ 9.7 | 4.9X ~ 31.5X | 46.8 ~ 7.4 | 7.35X ~ 47.25X | 31.4 ~ 4.8 | ||

0.5X | 174 | 3.5X ~ 22.5X | 65.6 ~ 10.2 | 4.37X ~ 28.12X | 57 ~ 8.8 | 5.25X ~ 33.75X | 44 ~ 6.8 | 7X ~ 45X | 32.8 ~ 5.1 | 10.5X ~ 67.5X | 22 ~ 3.4 | ||

1.5X | 57 | 10.5X ~ 67.5X | 21.8 ~ 3.4 | 13.12X ~ 84.37X | 19 ~ 2.9 | 15.75X ~ 101.25X | 14.6 ~ 2.2 | 21X ~ 135X | 10.9 ~ 1.7 | 31.5X ~ 202.5X | 7.3 ~ 1.1 |

Double click to read at full screen**Download Brochure (PDF)**

Double click to read at full screen**Download Manual (PDF)**